I can now input data from Labview to HP I managedto get communication between the equipment HP and LabView. Such error can occur if configuration of GPIB is not correct e. Next step I have tried before preparing the probestation was to set for one of the SMUs SMU1 a voltage value and measure the set value using a multimeter. I am new to the forum and also new to LabView.
|Date Added:||26 January 2015|
|File Size:||69.20 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Hp4145v 4 of Here should be what you are looking for. The only input commands I did were to set up teh channel definition and source setup.
Norbert Norbert CEO: Message 2 of Labveiw you have any experience with this? Message 7 of I will try nevertheless to find a solution. Message 8 of I am new to the forum and also new to LabView. A small update hp44145b the setup definition I use: Thanks for the feedback. Moreover I cannot see the trace of the sweep I expected a trace of 10 MA constant current and voltage ranging from 0 to 1 V not on the monitor of neither on the graph plot of the LabView VI interface. After detailing a bit more the process do you still think there might be a connection drop out between the GPIB interface and HP?
The data I want hp1445b represent is IV curves. I really appreciate all the feedback I can receive.
This is where Labveiw get error D 00 40 where after rading through the pages of the manual I got to the conclusion that D stands for data error, first two characters, xx, stand for X axis, the second set of 2 characters Y axis Message 9 of If i were you, i would contact KeySight formerly known as Agilent formerly known as HP what that message on the device is meaning.
Message 10 of This is on the B, right? I think there should be somewhere in the block diagrams of the HP B Example. I would like to ph4145b if there is any VI available for 2 terminal devices in this case on wafer diodes so that I can control directly the equipment from LabView and be able to read out the measured data directly from the HP via LabView? I will write Lavbiew, even though this product I believe to be discontinued, maybe they can offer me some support.
HP B Semiconductor Parameter Analyzer – Page 2 – HxC Floppy Drive Emulator
I will keep you posted with the outcome. I have no experience with LAbView so far.
During this time the multimeter which is connected to SMU1 output of the equipment register the voltage value. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
HxC Floppy Drive Emulator
The default error message for error 6 indicates an pabview in the GPIB communication. Error 6 I get it only when pressing “Take Measurement” button.
The VI instrument driver works. After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left to right when settinp a Sweep of Voltage labviww SMU1 from 0 to 1 V, pts. You can right click on the error cluster which shows error 6 and select “Explain Error”. Most Active Software Boards: Message 5 of What exactly is stopping bp4145b from doing this?